|Keywords||, , ,|
|Authors||Joshua M. Pearce|
|Designed in||United States|
|Hardware||CC BY-SA 4.0|
|Michigan Tech's Open Sustainability Technology Lab.
Wanted: Students to make a distributed future with solar-powered open-source 3-D printing and recycling.
Pearce Publications: Energy Conservation • Energy Policy • Industrial Symbiosis • Life Cycle Analysis • Materials Science • Open Source • Photovoltaic Systems • Solar Cells • Sustainable Development • Sustainability Education
- Michaels, R.E. and Pearce, J.M. (2017). 3-D printing open-source click-MUAC bands for identification of malnutrition. Public Health Nutrition, 20(11), pp. 2063-2066. https://doi.org/10.1017/S1368980017000726 open access
- Source files both SCAD and STL: https://www.youmagine.com/designs/muac-bands-for-identification-of-malnutrition
Color-coded 3-D printed click-MUAC bands. From left to right the bands have an inner circumference of 135 mm (green; three dimples), 125 mm (yellow; two dimples) and 115 mm (red; one dimple). (3-D, three dimensional; MUAC, mid-upper arm circumference)
|This Open Source Appropriate Technology has been designed but not yet tested — use at your own risk.|
|This Open Source Appropriate Technology has been modeled.|
|This Open Source Appropriate Technology has been prototyped.|
|This Open Source Appropriate Technology has been verified by MOST|
Objective: An effective method for the diagnosis of severe acute malnutrition is the measurement of the middle upper arm circumference (MUAC). Current methods to measure MUAC in a pre-hospital setting is through the use of measuring tape indicators, which require users to be trained in how to apply, adjust the tightness and read the device properly. This represents a challenge to using MUACs in many developing world contexts. This study explores the technical viability to overcome some of these challenges with conventional MUAC measurement methods using open source 3-D printable click-MUAC bands.
Methods: The dimensional accuracy of the open source 3-D printable click MUAC band was quantified with a digital micrometer to ensure reproducibility. The durability is evaluated by putting bands through a deformation test to simulate the use of the band in the field 500 times. The production costs are quantified using the mass of the filament and electricity consumed to manufacture each band.
Findings: The click MUAC bands are dimensionally accurate (inner circumferences of +/- 0.50 mm) and durable (surviving 500+ deformations with no notable residual deformation). The 3-D printable click MUAC bands are easier to use and provide a significant cost savings (92% to 97%) when compared to current MUAC measurement methods.
Conclusion: The open-source 3-D printed click-MUAC bands offer a viable alternative to the current methods of obtaining MUAC measurements.
See also[edit | edit source]
- Maximizing Returns for Public Funding of Medical Research with Open-source Hardware
- Open-Source Three-Dimensional Printable Infant Clubfoot Brace
- Economic Potential for Distributed Manufacturing of Adaptive Aids for Arthritis Patients in the U.S.
- Low-cost open source ultrasound-sensing based navigational support for visually impaired
- Parametric Nasopharyngeal Swab for Sampling COVID-19 and Other Respiratory Viruses: Open Source Design, SLA 3-D Printing and UV Curing System
- Open-Source Medical Hardware for Pandemics
- Additively Manufactured Parametric Universal Clip-System: An Open Source Approach for Aiding Personal Exposure Measurement in the Breathing Zone