|Part of||NREMT Skillset|
|Medical skill data|
|Subskill of||Airway Management|
|Acting roles||, , , ,|
|SDGs Sustainable Development Goals|
|License||CC BY-SA 4.0|
|Automatic translations||Français, Español, 中文, العربية, Русский, Kiswahili and others|
|Cite as GSTC (2021). "Rigid Catheter Suction". Appropedia. Retrieved 2021-08-1.|
Basic airway management and airway adjuncts will not protect against aspiration of gastric contents should vomiting occur - there is no replacement for effective suction. Airway suctioning should be performed whenever the patient's airway is potentially obstructed by secretions or foreign bodies. Field suction equipment consists of the following:
- Battery operated suction pump
- A vacuum canister for collection of fluids
- Flexible suction tubing
- Rigid suction tips for oropharyngeal suction, the most common of which are the Yankauer or tonsil tips.
- Flexible suction tips for nasal and stoma suction, commonly called "French catheters" which can come in several diameters measured in French or Fr
- Water for rinsing and flushing the suction tips
Both flexible and rigid suction tips can be "Whistle-tip" configuration meaning they have an opening close to where the tip attaches to the large diameter suction tubing in addition to the suction opening at the end. This "whistle" opening provides an alternate entry for air into the tip which means that suction at the end is very weak when this hole is open. Placing a finger over the opening directs all the suction to the tip, turning it "on". If the suction catheter does not have an opening, suction at the tip is continuous as long as the suction unit is on.
- Assemble. Attach the both vacuum side of the pump and the vacuum tubing to the fluid collection canister.
- Test. Turn the vacuum pump on, and test the tubing connections by bending the vacuum tubing to occlude it and listen for the sound of the pump to change indicating it is working harder. If this does not happen, check the airtightness of the seals between the vacuum pump, the canister and the tubing, and check that the canister lid is seated.
- Measure. Attach the appropriate tip for the type of suctioning being performed, and measure from the tip of the nose to the ear lobe on the patient to note how deep to insert the suction tip
- Suction. Insert the suction tip to the measured depth and then cover the whistle opening to suction only while withdrawing. For catheters without a whistle opening, the tubing may be kinked during insertion to prevent suction during insertion. Suction for no more than 15 seconds in an adult, 10 seconds in a child and 5 seconds in an infant
- Rinse. Suction tips may become clogged or jammed with debris, withdraw the tip and remove debris and flush with water as needed to maintain adequate suction.
- Oxygenate. Oxygen administration should be provided after suctioning.