3D Printed Otoscope
3D Printed Otoscope
Project developed by Amwallis
- An otoscope is simple device used in the diagnosis of ear, nose, and throat diseases. It is essentially a magnification lens with a light along with a speculum to help open orifices and direct the light. It is commonly used to examine the eardrum, which as the border separating the external ear canal from the middle ear, often has characteristics indicative of various diseases of the middle ear space.
- Picture of completed print using
Bill of Materials
- The main otoscope body weights 20 grams. The approximate cost of this part is $0.50. 87 min
- The battery cover weights 6 grams. The approximate cost of this part is $0.15. 19 min
- The switch cover weights 2 grams. The approximate cost of this part is $0.05. 4 min
- The circular base on which the lens assembly and the ear piece mounts weights 5 grams. The approximate cost of this part is $0.15. 22 min
- The ear piece weights 3 grams. The approximate cost of this part is $0.08. 22 min
- The lens assembly weights 3 grams. The approximate cost of this part is $0.08. 17 min
- A SMD pushbutton switch is used to activate the otoscope's light. The approximate cost of this part is $0.57.
- A white LED is used to shine light into the ear. The approximate cost of this part is $0.80.
- A 220 ohm resistor is used to limit current draw. The approximate cost of this part is $0.10.
- A 38mm diameter, 150mm focal length concave lens in used. The approximate cost of this part is $3.95.
- About half a meter of 22 gauge solid core wire is used. Cost is assumed to be negligible.
Total cost to manufacture is $6.43
Tools needed for fabrication of the OSAT
- MOST Delta RepRap or similar RepRap 3-D printer
- Depending on print quality, a knife may be needed to shave away plastic on the edges of the parts that come together.
- Super glue is needed to join the switch cover and the mount base to the main otoscope body.
- A soldering iron and solder is needed to securely attach the electrical components together
- Wire strippers and/or diagonals are needed to trim the wiring
- Rubber bands may be helpful to hold some of the parts together while the glue dries.
Skills and Knowledge Necessary to Make the OSAT
- You will need to know how to solder 
Technical Specifications and Assembly Instructions
- Provide directions for print/assembly - be detailed enough in your “how to” to ensure that someone could construct the device from your description. Consider the elegance of IKEA like instructions.
- Include print time estimate
- Include assembly time estimate
- Including drawings or pictures of the device at stage of assembly at minimum. (http://www.appropedia.org/Special:Upload)
- Consider video if appropriate
Common Problems and Solutions
- Include common mistakes/problems to avoid in building your OSAT and how to overcome them
- Assuming PLA plastic costing $0.025 per gram is used ($25 per 1kg spool), the otoscope can be built for $6.43
- A inexpensive home use otoscope costs approximately $30.00
- This results in a savings of $23.57 as the otoscope can be printed for 21% of the commercial equivalent