We continue to develop resources related to the COVID-19 pandemic. See COVID-19 initiatives on Appropedia for more information.
Field Ready Design - Nebuliser
Jump to navigation Jump to search
Project developed by [User:fordahl|Joey Fordahl]
- Sunil works as a Biomedical Engineer at Grande International Hospital, Kathmandu. A piece of equipment that he repeatedly sees in the engineering department is the nebuliser compressor. A nebuliser works by using compressed air to convert liquid medicine to an aerosol that can be inhaled by the patient. However, the plastic connector which attaches the medication tube to the compressor keeps breaking 
- Once a new hole is drilled through the nebuliser's exterior using a 12 mm drill bit, my connector will slide into the hole and be fastened with a 3d-printed nut. My connector was designed using FreeCAD from the dimensions provided by the site. I used the threaded portion of an M12-1.75 bolt that I pulled from Mcmaster.com. The nut stl file is also from Mcmaster.
Bill of Materials
- 3D-printed 12mm-1.75 flange nut (Mcmaster STEP file)
Tools needed for fabrication of the OSAT
- MOST Delta RepRap or similar RepRap 3-D printer
- 1.75 mm PLA
Skills and Knowledge Necessary to Make the OSAT
- All you need is a 3D printer and slicing software for the STL's provided
- If you would like to modify the design, you can download FreeCAD 
Technical Specifications and Assembly Instructions
- Both the connector and nut were printed in under 1 hour
- The connector file should be printed with Support Type: Touching Buildplate (60 degree overhang)
- Both files were printed at 0.1 mm layer height, but could be printed at .25 mm if desired.